A Compton suppression system is used to reduce the contribution of scattered gamma-rays that originate within the HPGe detector to the gamma-ray spectrum. The HPGe detector is surrounded by an assembly of guard detectors, usually NaI(T1). The HPGe and NaI(T1) detectors are operated in anti-coincidence mode. The NaI(T1) guard detector detects the photons that Compton scatter within, and subsequently escape from the HPGe detector. Since these photons are correlated with the partial energy deposition within the detector, much of the resulting Compton continuum can be subtracted from the spectrum reducing the unwanted background in gamma-ray spectra. A commercially available Compton suppression spectrometer (CSS) was purchased from Canberra Industries and tested at the Radiation Science and Engineering Center at Penn State University. The PSU-CSS includes a reverse bias HPGe detector, four annulus NaI(T1) detectors, a NaI(T1) plug detector, detector shields, data acquisition electronics, and a data processing computer. The HPGe detector is n-type with 54% relative efficiency. The guard detectors form an annulus with 9-inch diameter and 9-inch height, and have a plug detector that goes into/out of the annulus with the help of a special lift apparatus to raise/lower. The detector assembly is placed in a shielding cave. State-of-the-art electronics and software are used. The system was tested using standard sources, neutron activated NIST SRM sample and Dendrochronologically Dated Tree Ring samples. The PSU-CSS dramatically improved the peak-to-Compton ratio, up to 1000:1 for the 137Cs source.
All Science Journal Classification (ASJC) codes
- Analytical Chemistry
- Nuclear Energy and Engineering
- Radiology Nuclear Medicine and imaging
- Public Health, Environmental and Occupational Health
- Health, Toxicology and Mutagenesis