Vitek 2 (bioMérieux Inc., Durham, NC) is a widely used commercial antimicrobial susceptibility test system. We compared the MIC results obtained using the Vitek 2 AST-GN69 and AST-XN06 cards to those obtained by CLSI broth microdilution (BMD) for 255 isolates of Enterobacteriaceae, including 25 isolates of carbapenem-resistant Enterobacteriaceae. In total, 25 antimicrobial agents were examined. For 10 agents, the MIC data were evaluated using two sets of breakpoints: (i) the Vitek 2 breakpoints, which utilized the 2009 FDA breakpoints at the time of the study and are equivalent to the 2009 CLSI M100-S19 breakpoints, and (ii) the 2014 CLSI M100-S24 breakpoints. There was an overall 98.7% essential agreement (EA). The categorical agreement was 95.5% (CA) using the Vitek 2 breakpoints and 95.7% using the CLSI breakpoints. There was 1 very major error (VME) (0.05%) observed using the Vitek 2 breakpoints (cefazolin) and 8 VMEs (0.5%) using the CLSI breakpoints (2 each for aztreonam, cefepime, and ceftriaxone, and 1 for cefazolin and ceftazidime). Fifteen major errors (MEs) (0.4%) were noted using the Vitek 2 breakpoints and 8 (0.5%) using the CLSI breakpoints. Overall, the Vitek 2 performance was comparable to that of BMD for testing a limited number of Enterobacteriaceae commonly isolated by clinical laboratories. Ongoing studies are warranted to assess performance in isolates with emerging resistance.
All Science Journal Classification (ASJC) codes
- Microbiology (medical)